The good functioning of the native, or repaired valve depends on coaptation capacity of its leaflets. Annuloplasty rings have been used since 1968 to remodel the shape, correct the dilatation, and consolidate the repair of heart valves by improving coaptation of the valve's leaflets during systole.
The further understanding of the three-dimensional geometry of the native valve annulus during the normal cardiac cycle lead to the evolution of the rings from the first rigid, planar stainless steel-based Carpentier ring to the fexible Duran ring and to the more recent Cosgrove-Edwards bands. All these devices, which actually fulfil the requirements of the application on adults, do not allow the further growth of the native annulus if implanted in children.
Indeed, implantation in growing hearts could result in a stenotic effect which could worsen with time. For this reason, these current traditional rings do not exist in pediatric sizes (under 26).
A NEW CONCEPT IN ANNULOPLASTY VALVE REPAIR
These advanced devices, an example of which is shown in Figure 1, do not work like traditional rings. Indeed, the device does not provide a mechanical support, except in the first short period following implantation, but induces the creation of endogenous fibrous tissue which will take on this function subsequently. Moreover, it was designed to facilitate its implantation.
The device is introduced within the native annulus and simply secured to the anterior and posterior trigones by knotting the integrated monofilament sutures equipped with crimped stainless steel atraumatic needles.
This intra-annular implantation prevents the device from being in contact with blood. This avoids thromboembolic complications which require systemic anticoagulation therapies until the endocardium covers the traditional rings sutured on the native annulus. In addition, Bioring's device is solely composed of materials that can be used in presence of endocarditis.
In order to preserve the native annulus growth potential in children, the non-resorbable suture that is incorporated in the polydioxanone segment, in order to prevent further degenerative dilatation of the annulus in adults, is cut in pediatric sizes of the ring under 26.
Six months or more following implantation, a fibrous autologous tissue was developed in all dissected hearts, in replacement of the implanted polydioxanone device, resorbed by hydrolysis. Moreover, it was discovered that the fibrous tissue continues on growing for a few months after complete resorption of the device. The newly grown fibrous tissue remodells the annulus and prevents it against redilatative stretch without interfering with its dynamical physiological motion.
CLINICAL STUDIES AND TRIALS
The preliminary clinical study that focussed on pediatric population gave positive outcomes and confirmed the results obtained in the animal trials. This demonstrated the biodegradable device's main feature for pediatric applications: the preservation of the growth potential of the native annulus4.
The benefits of the valve repair with Bioring's biodegradable annuloplasty device were assessed in terms of NYHA Grade for the 105 patients in whom the follow-up was complete. They were immediately tested post-op and monitored at 6, and 12 months. This evaluation was utilized as it is one of the most commonly used scoring systems worldwide. These benefits are represented in Figure 2. It can be seen that all patients fully recovered and are presently scored Grade 0 or Grade 1. Moreover, in this study it is also demonstrated that the degradation of the polymeric segment of the device occurs without any observable consequences for the patient.
In conclusion, the Kalangos® ring is a new biodegradable device that has proven to be safe and efficient. Moreover it is the only commercial device available in pediatric sizes for correcting, remodelling the shape, reinforcing, and restoring the function of the valves' annuli, that preserves the native annulus growth potential in children. The tissue-growth properties associated with the intra-tissular implantation of this device introduce a completely new concept in annuloplasty.
BIORING, founded in 2000, develops and commercializes biodegradable medical devices that lower surgical risk and reduce post-surgical complications. The first device in the series, The Kalangos® ring, received the CE Mark in March 2005. BIORING is currently working on obtaining the FDA approval for this device.
1. Chachques JC, Acar C, Latremouille C, et al. 'Absorbable rings for pediatric valvuloplasty'. Circulation 1990; 82 Suppl IV: 82-8.
2. Miyamura H, Eguchi S, Watanebe H, et al.. 'Total circular annuloplasty with absorbable suture for the repair of the atrioventricular valve regurgitation in atrioventricular septal defect'. J Thorac Cardiovasc Surg 1994; 107: 1428-31.
3. Myers JL, Campbell DB, Waldhausen JA. 'The use of absorbable monofilament polydioxanone suture in pediatric cardiovascular operations'. J Thorac Cardiovasc Surg 1986; 92: 112-8.
4. Albanese S, Carotti A, Lacobelli R, et al. 'Mitral valve repair supported by the Kalangos ring in a pediatric population': 4th World Congress of Pediatric Cardiology and Cardiac Surgery. Buenos Aires, Argentina, September 2005.